In past ramblings on particle size analysis we have touched on shakers, ASTM standards, sieve checking, separating small particles, inhibitors like static charges and how to get sieve tests done. A couple of things that we rarely addressed.Read More
CSC Scientific Blog
Do You Re-Certify? Compare with a Master Stack? Check with Calibration Samples?
What Does Recertification Give?
The ASTM E-11 committee has done a great job of establishing three levels of testing. These levels show the probability of a sieve’s mesh to be within the permissible variations. These variations relate to the size of openings in wire-cloth used for test sieves.Read More
Topics: Sieve Shakers, Mid-Point Sieves, ASTM, Particle Size Analysis, Sieve Calibration, Sieve Certification, Sieve Testing, Sieving Process, Quiet sieve shakers, RoTap, sieving, test sieve equipment, sieve mesh
Quiet Sieve Shakers
Recently in a quiet, reflective moment, I recalled my first work with sieve shakers. Horizontal motion with tapping was the basic shaker design. The Ro-Tap® was King, and there were few alternatives. Different shakers for special applications, such as the Mary Jane and one that was hung from the ceiling, were the exception.Read More
Visitors to our web pages often arrive with the question, “What is the Function of a Sieve Shaker?”
The simple answer is “to expose the particles in a sample to all the openings in each sieve in a stack”. A sieve stack is the result of fitting each sieve to be used in a given particle size analysis into the one above. The sieve with the largest mesh holes is at the top with each subsequent sieve of a tighter mesh size than the one above it.